DSW-Series IP65 Pressure Transducer by Proportion-Air
Proportion-Air manufactures the DSW-Series IP65 pressure transducer for industrial marekts. The DSW-Series is rated NEMA4/IP65, but it is electrically the same as the DSB-Series industrial pressure transducer. This means the DSW offers great accuracy and repeatability for industrial environments. The IP65 version is housed in a slightly larger canister that is sealed against fluid spray from any direction.
The DSW IP65 pressure transducer accurately measures the pressure of various media with an electrical output signal that is proportional to the full calibrated pressure range. For example, a 0-150 PSI calibration with a 0-10 VDC output would read 5 VDC at 75 PSI.
What is IP65?
This Ingress Protection (IP) rating = Totally protected against dust & against low pressure jets of water from all directions.
This IP65 pressure transducer can be used as a standalone transducer or as 2nd loop feedback in a Proportion-Air 2-loop pressure control assembly. The DSW IP65 pressure transducer can be configured with a variety of connectors or with flying leads for the best fit into your application.
DSW IP65 Pressure Transducer
- Pressure Range: Vacuum thru 175 PSI (12 Bar)
- Accuracy: ±0.2% F.S.
- Repeatability: ±0.02% F.S.
- Media: Non-Corrosive Gases
- Weight: 11.2 oz. (318g)
- Protection Rating: NEMA4/IP65
- Response Time: 100 Microseconds
- Port Size: 1/8″ & 1/4″ NPT
- Durability: Immune to shock & vibration (up to 100 Gs)
- Operating Temps: 32-158°F (0-70°C)
- Output Signals: Voltage or Current
What are customers are saying
What an excellent process you have for specifying parts for your customers. I am very impressed.
I met with Bill today...He was impressed by your knowledge and application of your products and wanted me to thank you for your excellent support.
It is really a pleasure to work with Proportion-Air. You cannot imagine what kind of problems I have with some manufacturers. Thank you very much!
I'm astonished for the quickest answer ever!
I'm happy to report that the trial on the QPV1 has been going well. I'm quite impressed by the performance of the unit. I'm also very impressed by the service that I received from Nick Archer which I think merits specific attention. Nick has been extremely responsive to my emails and questions. He has been patient, has kept his word, and has always replied to me quickly with an answer or a reply to my questions. Nick has provided me with what I, and certainly others in my group, consider to be a perfect example of top notch high quality customer service.